I opened the Configure Standard Waveform block and this is what it looks like:. Why is the output voltage twice the value I set? I have the programming guide for the generator, and I understand the idea of the command string that is fed to it, but tinkering around with this might be a waste of time for me since the IVI drivers work ok. You can also troubleshoot and make manual function calls to your instrument to ensure that it is working as expected before moving into automation. How do I disable this?
|Date Added:||7 January 2010|
|File Size:||15.40 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Do the same for the burst phase and burst period terminals on the left side and bottom of the Configure Burst node. Log in to follow, share, and participate in this community. Rate this document Select a Rating 1 – Poor 2 3 4 5 – Excellent. So it would have to be part of the driver code. Instument is the A. Most Active Software Boards: How agilemt I disable this?
Agilent 33XXX Series Signal Generator
How do I sweep in both directions. I am outputting morlet wavelet from A, but the signal is outputting continuously. For this example, focus on the section between the Configure Arbitrary Waveform node and the Enable Output node.
Application Programming Interface API To help accelerate software development, all instrument drivers follow a consistent programming flow of: I am using the Keysight B series waveform generator to frequency sweep a pulse waveform.
Message 7 of 7.
When writing an instrument control application for a serial instrument, you communicate with a COM port rather than the serial instrument connected to the COM port. Back to Top 2.
Waveform Generators | Keysight Community
The 33xxx driver takes ms to change, and has a strange burst signal in between. Thanks for the suggestion Mateus, but I am trying to automate this process as best I can. I have the programming guide for the generator, and I understand the idea of the command string that is fed to it, but tinkering around with this might be a waste of time for me since the IVI drivers work ok.
The NI Instrument Driver Network houses thousands of instrument drivers for automating third-party labvirw.
TestStand, a ready-to-run test management software, is designed to help you develop, execute, and deploy automated test and validation systems faster. However, it is still showing twice the set voltage. If you select the device, you can see additional properties, such as the device name, and any software or drivers that are installed on your system to support this device. I am currently using the A and am encountering a strange error.
The instrument is then detected. General Use Programming and Remote Communications clear selected. The instrument is now configured. Place Controls onto the Panel using the Unplaced items box. You will need to rewrite this part of the driver keep it in your project library and leave the actual driver alone if this is how it works.
You can configure the Configure Burst node with several inputs. Download Agilent Series Instrument Driver. It was working just fine earlier.